1, image measuring instrument should be placed in a clean and dry room (room temperature 20 - 2 Deg. C, humidity 40%-60%), avoid optical surface fouling, metal parts rust, dust debris falling into the motion guide, influence of instrument performance.
2, after the use of image measuring instruments, the work surface should be cleaned at any time, it is best to cover the dust sheath.
3. The transmission mechanism and the moving guide rail of the image measuring instrument should be lubricated regularly, so that the movement of the mechanism is smooth and the state of use is kept good.
4, the table glass and paint surface dirty, can be wiped with neutral detergent and clean water. Never wipe the paint surface with organic solvent, otherwise it will tarnish the surface of the paint.
5, the image measuring instrument LED light source has a long service life, but when the bulb is burned out, please notify the manufacturer, replaced by professionals for you.
6, image measuring instrument precision components, such as imaging system, working platform, optical rule and Z axis transmission mechanism are precision adjustment, all adjusting screw and fastening screws were fixed, customers do not disassemble, if you have any questions please inform the supplier to solve.
7. The software has made accurate compensation to the error of the worktable and the optical ruler. Don't change it automatically, otherwise, it will produce wrong measurement result.
8. All electrical connectors of image measuring instruments should not be pulled out. If it has been pulled out, it must be inserted and tightened according to the mark correctly. Incorrect insertion may affect the function of the image measuring instrument and even damage the whole system.